Clinical Evaluation of the 3nethra Aberro Handheld Autorefractometer
Purpose: To evaluate the 3nethra aberro auto refractometer device as an alternative tool for quick and reliable measurement of refractive errors and to compare it with the gold standard subjective refractive error measurement.
Methods: Refractive errors were measured using both subjective refraction and the 3nethra aberro handheld autorefractometer. The refractive measurements were converted into equivalent vector notations of spherical equivalent and Jackson cross-cylinder measurements J0 & J45. The resultant power vectors were compared with subjective measurements.
Results: This clinical study comprised 60 subjects (22 male and 38 female; with a mean age of 34 ± 16 years). Data, when compared with the subjective refraction measurements, resulted in 90% of power vectors values in both left and right eyes being the same in the 3nethra aberro handheld autorefractometer and the subjective measurement. The refractive error measurements also had an agreement of 70% and 90% when the range of diopter was between ±0.25 and ±0.5D, respectively. When the Bland-Altman’s plot analysis was performed, about 98% of data lied within the ±2 standard deviation variation. An average correlation between the two methods of error measurement was 0.74, and the paired t-test showed P > 0.05 for all the power vectors except for the spherical equivalent in the right eye.
Conclusion: The 90% agreement between the error measurements done by two methods indicates that the 3nethra aberro handheld autorefractometer can function as an alternative for the time-consuming subjective refractive error measurement.
Aberro Autorefractometer, Non-mydriatic, Pupil Diameter, Refractive Index, Subjective Refractive Error, Vision, Wavefront Technology
1. Resnikoff S, Pascolini D, Mariotti SP, Pokharel GP. Global magnitude of visual impairment caused by uncorrected refractive errors in 2004. Bull World Health Organ 2008;86:63–70.
2. Bao J, Le R, Wu J, Shen Y, Lu F, He JC. Higherorder wavefront aberrations for populations of young emmetropes and myopes. J Optom 2009;2:51–58.
3. Bennett JR, Stalboerger GM, Hodge DO, Schornack MM. Comparison of refractive assessment by wavefront aberrometry, autorefraction, and subjective refraction. J Optom 2015;8:109–115.
4. Lebow KA, Campbell CE. A comparison of a traditional and wavefront autorefraction. Optom Vis Sci 2014;91:1191–1198.
5. Salmon TO, van de Pol C. Evaluation of a clinical aberrometer for lower-order accuracy and repeatability, higher-order repeatability, and instrument myopia. Optometry 2005;76:461–472.
6. Khan MS, Humayun S, Fawad A, Ishaq M, Arzoo S, Mashhadi F. Effect of wavefront optimized LASIK on higher order aberrations in myopic patients. Pak J Med Sci 2015;31:1223–12236.
7. Wei Z, Jianfeng L, Funian L, Zhile W. Study on wave front fitting using Zernike polynomials. OT 2005;31:675–678.
8. Liang J. Wavefront technology for vision and ophthalmology. Aberration-free refractive surgery. Berlin, Heidelberg: Springer; 2004. p. 25–47.
9. Bruce AS, Catania LJ. Clinical applications of wavefront refraction. Optom Vis Sci 2014;91:1278–1286.
10. Harris WF. The Jackson cross-cylinder. Part 1: Properties. Afr Vision Eye Health 2007;66:41–55.
11. Thibos LN, Wheeler W, Horner D. Power vectors: An application of Fourier analysis to the description and statistical analysis of refractive error. Optom Vis Sci 1997;74:367–375.
12. Durr NJ, Dave SR, Lim D, Joseph S, Ravilla TD, Lage E. Quality of eyeglass prescriptions from a low-cost wavefront autorefractor evaluated in rural India: Results of a 708-participant field study. BMJ Open Ophthalmol 2019;4:e000225.
13. McGinnigle S, Naroo SA, Eperjesi F. Evaluation of the autorefraction function of the Nidek OPD-Scan III. Clin Exp Optom 2014;97:160–163.
14. Shneor E, Millodot M, Avraham O, Amar S, Gordon-Shaag A. Clinical evaluation of the L80 autorefractometer. Clin Exp Optom 2012;95:66–71.
15. Mello GR, Rocha KM, Santhiago MR, Smadja D, Krueger RR. Applications of wavefront technology. J Cataract Refract Surg 2012;38:1671–1683.
16. Maeda N. Wavefront technology in ophthalmology. Curr Opin Ophthalmol 2001;12:294–299.
17. Durr NJ, Dave SR, Vera-Diaz FA, Lim D, Dorronsoro C, Marcos S, et al. Design and clinical evaluation of a handheld wavefront autorefractor. Optom Vis Sci 2015;92:1140–1147.
18. Rubio M, Hernández CS, Seco E, Perez-Merino P, Casares I, Dave SR, et al. Validation of an affordable handheld wavefront autorefractor. Optom Vis Sci 2019;96:726–732.